led to Circul8 Max, the next generation of at-home DVT prevention.
The pump will inflate each leg cuff to a preset pressure of 55 mmhg and deflate once the pressure is reached. The cycles are repeated on each unit until the power is turned off.
The device housing may emit minor warmth during use or during charging, this is not cause for concern. If the device becomes HOT to touch, discontinue use and contact us by phone 888-767-0015.
Yes, our products are approved by the TSA.
During normal operation, the device can be turned into night mode by quickly pressing the Power Button Three Times. The light intensity will reduce on the power button, and the LED display will turn OFF.To exit night mode, quickly press the Power Button Three Times. If the power is turned OFF for any reason, the device will automatically reset and start in normal operation mode.
While powered off, it takes 4 hours to charge from depleted state.
While powered off, it takes approximately 4 hours to charge from depleted state.
When powered on and continuously operating, the battery life from full-capacity to total depletion can last up to 20 hours. Altitude, hot, and/or cold temperatures can affect battery life. Avoid extreme temperatures.
Wear it anytime you are sitting or sleeping more than one hour. The greatest risk occurs when you are not moving, so feel free to use it as much as you like.
While Circul8 Max does allow for the device to operate while it is charging, we advise against this. Operating the device(s) while charging may result in inefficient and longer charging time and could result in harm to the battery. Use only the charger that is included within your Circul8 Max package.
No, the device is yours to keep. There is no need to return the device. In the event a device needs to be returned, our Patient Care Coordinator will contact you.
Please continue wearing the functioning device on the affected leg (surgery leg) and call 888-963-6265 to see if we can help troubleshoot the problem. If the troubleshooting is unsuccessful, the patient/caregiver may pick up a new one at the clinic, or we will ship a new device to you.
Wipe down with a damp cloth, mild soap and water. Do not submerge the device in water. Do not expose the housing box or any of the mechanical / electrical components to water.
STEP 1.) If you feel the device isn’t inflating properly, first check that the device is properly secured to your leg. Proper usage should have the entire garment of the device fitting snugly around your leg. To ensure proper fit, follow the instructions in the quick start guide and/or the instructions for use.
STEP 2.) Upon confirming proper fit, if you feel that the device is still inflating improperly, remove the device from your leg, connect it to the power supply cord included within your Circul8 Max package and allow the device to charge for a minimum of 3 hours. Prior to charging, please inspect the charging cord for defects, according to the steps listed in “What should I do if my device does not appear to be charging when connected to the power supply and/or will not hold a charge after it has charged?“. Once you have allowed the device to recharge, reapply the device to your leg according to the directions listed in Step 1, power on and allow the device to operate for 5 minutes. If the inflation issues still persist, please contact customer service to request a replacement.
Battery icon: Remains illuminated at all times during operation.
Solid Green All Three Bars: 100% battery life remaining.
Solid Green Two Bars: 68% to 100% battery life remaining, when charging in use or sleep mode, third bar pulses until fully charged.
Solid Green One Bar: 33% battery life remaining, second and third bar pulses alternately until 100% charged when charging in use or sleep mode.
Flashing Yellow Bar: Power Button will flash Yellow, and “BL” quickly flashes on the LED display. The battery will last 30- 60 minutes or less.
Flashing RED Bar: Battery will last five minutes or less. “BC” quickly flashes on the LED display until auto connected while the device is in use.If lights continue to flash: make sure the wrap is snug around the calf. Turn the Circul8 Max unit off, and then power the Circul8 Max unit back on. If the unit continues to flash lights and the default pressure is not reached, please call 1-888-767-0015.
STEP 1.) Please look at the charging cord and the metal connector pins that plug into the wall outlet. If you see any rips or cuts on the cord or if the metal connector pins are loose or bent, it’s likely that your charging cord is broken and needs to be replaced. In the event of a broken charger, please contact customer service for a replacement.
STEP 2.) Once the power supply cord has been inspected and it is determined that there are no defects, connect your Circul8 Max device(s) to the power supply cord. Once the device is connected and plugged into a functional wall outlet go to step 2a and 2b.
STEP 2a) The Red Indicator Light located around the Power Button illuminates: The device is actively charging. Allow the device to charge for a minimum of 3 hours. If the device continues to have trouble maintaining power after it has sufficiently charged, please contact our customer support specialist to request a replacement.
STEP 2b) The Red Indicator Light located around the Power Button does NOT illuminate: The device is NOT receiving a charge and there may be an issue with the power supply cord or with the wall outlet to which it is connected. The power supply cord included within your Circul8 Max box has 2 Connector Pin’s located at the end of the cord. To troubleshoot, try each connector pin with the device. If the red light still does not illuminate, attempt to plug the power supply into an alternative wall outlet.
After completing these steps, if you continue to experience charging-related issues, please contact our customer support specialist to request a replacement.
Power the device off, allow it to rest momentarily, then attempt to power the device on again. If the display lights still do not appear, remove the device from your leg, connect it to the power supply cord included within your Circul8 Max package and allow the device to charge for a minimum of 3 hours. Prior to charging, please inspect the charging cord for defects, according to the steps listed in “What should I do if my device does not appear to be charging when connected to the power supply and/or will not hold a charge after it has charged? “. Once the device has been charged sufficiently, reapply the device to your leg and apply proper fit according to the directions listed in Step 1 of section “What if the device is not inflating properly?”. Once it is secured to your leg, power on the device. If the display lights still do not appear, please contact our customer support specialists to request a replacement.